A positive displacement flow meter is a type of flow meter that requires fluid to mechanically displace components in the meter in order for flow measurement. Positive displacement (PD) flow meters measure the volumetric flow rate of a moving fluid or gas by dividing the media into fixed, metered volumes (finite increments or volumes of the fluid). A basic analogy would be holding a bucket below a tap, filling it to a set level, then quickly replacing it with another bucket and timing the rate at which the buckets are filled (or the total number of buckets for the “totalized” flow). With appropriate pressure and temperature compensation, the mass flow rate can be accurately determined.
Positive Displace Flow meter Overview
Complete sensor systems provide additional capabilities such as an integral display and/or user interface. For both types of positive displacement flow meters, performance specifications include the minimum and maximum measurable flow rate, operating pressure, temperature range, maximum allowable material viscosity, connection size, and percent accuracy (typically as a percentage of actual reading, not full scale).
Positive Displace Flow meters Principle
These devices consist of a chamber(s) that obstructs the media flow and a rotating or reciprocating mechanism that allows the passage of fixed-volume amounts. The number of parcels that pass through the chamber determines the media volume. The rate of revolution or reciprocation determines the flow rate. There are two basic types of positive displacement flow meters. Sensor-only systems or transducers are switch-like devices that provide electronic outputs for processors, controllers, or data acquisition systems.
|Volume per Revolution
- vortex Flow meter manual
- Electromagnetic Flowmete catalog
- inline vortex Flow meter 1“ inch drawing
- inline vortex Flow meter 1.5“ inch drawing
- RS485 Protocol
- Hart Protocol